CUT OUTPUT POWER STATED LOAD PURE CUT 300 W 300OHMS BLEND 150 W 300OHMS BLEND 75 W 300OHMS BLEND 90 W 300OHMS
COAGULATION SPRAY COAG. 120 W 300OHMS POINT COAG. 80 W 300OHMS
BIPOLAR (Frequency 500 kHz) BIPLOAR COAG. 80 W 100OHMS BIPLOAR CUT. 120 W 300OHMS
CONTROL INPUTS: MONOPOLAR Dual pedal Footswitch and / or Dual control Handswitch.
BIPOLAR Single pedal Footswitch / Bipolar forceps.
AUDIBLE SIGNALS: Separate audio tones for CUT / COAG. / BIPOLAR commands. Alternating audio tone for patient plate fault. (If under command)
POWER AC line Voltage 200 Volts to 250 Volts AC RMS, 50 Hz single phase. Machine output unaffected in the specified range, hence suitable to use without additional voltage stabilizer.
SIZE : Width 415mm, Height 160mm, Depth 385mm
WEIGHT : 8 Kgs.
ACE - 300 - The most progressive Electrosurgical generator
Its unique State - of - The Art technology offers
Monopoloar & Bipolar output
Spray coagulation & precise pinpoint coagulation
Pure CUT mode and three other modes of Blending at finger tip.
Plus numerous safty features. truly a sugaon's choice
Spray coagulation with safe isolation quickly controls bleeding with minimal tissue destruction. We call it ‘Super Spray’. That’s because it provides superior control of bleeding with less depth of tissue necrosis and minimal destruction. Spray coagulation can be used to give rapid fulguration for virtually all types of procedures from delicate micro-surgery to demanding oncology and thoracic surgery.
Precise pinpoint coagulation is also achieved in monopolar mode and provides precise control of localized bleeding. So it is ideal for use in a small, defined area.
Pure cutting mode plus three other selected BLEND modes (mixing of CUT & COAG effect) for superior blending options. The CUT mode offers PURE CUT for smooth, clean cutting.
MONOPOLAR a) Footswich : CUT/COAG. Pedals. b) Pencil swich : Cut/COAG. Button Appropriate LED’s turn on with respective command input. a) and b) may be simultaneously connected to the machine and any one can issue command audio tone, to identify the CUT / COAG. command.
Interlocks: Simultaneous issue of foot switch or pedal switch (or CUT from Footswitch and COAG. from Pencil switch) will cancel the commands, resulting in no audio tone, command LED and no power output. This is a safety feature.
If Alarm LED is ON, (i,e. patient lead fault), then monopolar commands from Monopolar Footswitch / Handswitch are disregraded, i. e. CUT / COAG. LED’s cannot turn on. However a dual audio tone only is issued on any monopolar command. No surgical power output can occur.
BIPOLAR: a) Footswitch (single pedal) Command can be issued from Bipolar Footswitch if the panel indicator is FS in the bipolar block. Issue of the command is indicated by LED as well as an audio tone. Precedence and Interlocks: Bipolar has precedence over monopolar that means if bipolar & monopolar are issued simultaneously, then bipolar is executed disregarding monopolar. PPF (Patient Plate Fault). disconnection of one or both leads of patient plate circuit will turn on ALARM LED. In this case, any monopolar command will produce 2 tone audio, no command LED, no issue of monopolar command & no monopolar power. Bipolar block remains unaffected and is functional in spite of ALARM LED ON as it is not related to patient plate of monopolar generator.